Mill/Lathe Digital Read Out

This page describes how to use the digital read outs for the manual mills and lathes.

Manual Mills
The manual mills have 2 kinds of digital readout system, the 100S and 200S readouts from Acu-Rite.

Setup
The system can be turned on by a switch on the back of the display. Take care to remember that after 90 minutes a decimal point will roll across the screen signalling that it is idle, which can be cancelled by pressing any button.

Zeroing
Before you begin you need to perform the zeroing process. The details of this process are posted on the wall behind the manual mills.

Once the mill is in the correct position the datum can be set by ensuring you are in absolute mode by pressing the "abs/incr" button. You then press the "zero" button on the right side of the screen that corresponds to the correct axis. This allows you to enter the value the mill is at relative to the datum and is confirmed by pressing "Enter". It is important to note the radius of your zeroing tool, and to make sure you are using the correct sign relative to what side of the part the mill is on. It is always worth it to visually check your zeroes as well.

By switching to incremental mode using the "Abs/Incr" button to can rezero the DRO without removing your datum. This allows you to measure lengths without keeping track f start and end positions. When you are finished you can see the position relative to the datum by again pressing the ABS/Incr button.

Units
The can use either inches or millimeter units. Ensure you are using the correct units by pressing the "mm" button and checking the readout to tell you what the system is using.

Manual Lathes
The manual lathes's use NMS300 digital readouts. This guide will cover the basic functionality of these displays for most applications.

The Display: The display automatically turns on when the device is in use.



Zero/Set Mode: There are 2 main modes in which one can define an axis on the NMS300 Displays. These are the zero and set modes. A user can toggle between these modes by pushing the zero/set button. The selected mode will be displayed on the screen in the bottom right corner. Zero Mode: When in zero mode, to zero an axis, a user must simply move the lathe along the axis to the location in which they would like to define as zero. Then. the user must select the arrow button immediately to the right of that displayed axis. This will zero only that axis.

Set Mode: Set mode allows users to define the current location of an axis. First, move the mill to the desired location. Next, select the arrow next to the axis to be defined. Lastly, type in a numeric value to set that axis location to and hit enter. It is also possible to zero an axis in set mode. To do this, simply hold the arrow next to the axis to be zeroed for three seconds.

Undo Function: The undo function allows users to undo their last input into the display. To undo a previous input, such as zeroing an axis, press the undo button.

Centre Find Function: The centre find function allows user to divide an axis value in half. First, push the 1/2 button and then select the arrow next to the axis to be halved.

Unit Change: To switch in and mm, select the in/mm button. The current units will be displayed at the bottom left of the screen.