Trunk Bending Apparatus

Trunk stiffness and stability can be related to the posture and alignment of the human body. This project continues on the evolution of the Twister, a Trunk Stiffness Bending Device, that is intended to measure and acquire data of axial and torsional stiffness in core muscles. Then, postural performance can be linked to measured torso stiffness to examine muscle reaction patterns with the consideration of age, height, weight, and other factors.

Background
In the 1980's, Dr. Victor Gurfinkel, medical researcher at Oregon Health & Science University, developed the Twister to measure torsional muscle tone as well as muscular responses as the apparatus twisted the subject. The body sections considered by the Twister are the neck, trunk, and waist. Then, Dr. Gurfinkel devoted the apparatus to Dr. Rajal Cohen at University of Idaho, Psychology & Communication Studies Department, upon his retirement.

The developers of the Twister seek to study the psychological mechanism caused by the change in length of muscles crossing the selected region as a result of the device twisting that region. The resulting resistive torque is determined to calibrate the tonic control of all muscles subjected to changes in length. Based on peek-to-peek magnitude, phase of resistive torque, and tonic electromyogram modulation, the results being acquired show magnitude and dynamic control of muscle activity. Moreover, Fay Horak, another developer of the Twister at Oregon Health & Science University, believes that the Twister will help answering how the brain control normal postural tone, and how abnormal tone can be improved with treatment [].

Problem Definition
The previous team assembled and modified the structure to be relevant for the trunk bending experiments. They created an interface software to collect data, upgraded the hardware, and built harnesses. The objective is to bring the machine. into an operational state by using higher quality parts, completing circuitry, making the machine more user-friendly, adding safety features, and testing and collecting data from the final product

Initial Project State & Goals
Proceeding the project from the point reached by the previous team:
 * New string potentiometers are needed to withstand the weight applied and be durable. As new potentiometers are going to be used, the circuitry to the power supply and computer needs to be set up differently and the software interface created last year needs to be reprogrammed.
 * Strain gauges need to be programmed through the torque sensor.
 * New timing belt and pulleys are to be replaced for the bottom potentiometer.
 * The wiring of power supply seems to be done incorrectly therefore, the power supply has to be rewired correctly to the computer and all sensors.
 * The current neck bar has only one universal joint and as a result, the subjects neck has to bend along with the movement of the neck bar. To allow movement while the subject's neck is still perpendicular to the ground, two universal joints are to be installed so that one cancels the bending angle of the other.
 * Torque sensors need to be reconnected correctly to the power supply then, the its sensor has to be reprogrammed to the software interface created last year.
 * The current attachment bar system is not practical and can hardly be placed accurately. Therefore, a roller system is going to be added to the lateral bars so that the bars remain horizontal and can be slid up and down easily.
 * The DC motor has to be connected to the circuit and programmed with the software interface.
 * An emergency stop is to be added to the system for safety.
 * The current system does not have a physical safety stop when the platform rotates. Therefore, a physical stop is to be added to the system to have the platform stops at a certain degree while still achieving maximum rotation required.

Turntable
The current set-up of the potentiometer at turntable has a 1:1 gear ratio however, in order to have the turntable function as needed, the set-up has to have a 6:1 gear ratio. For this to be achieved, the pulley attached to the potentiometer as well as the timing belt attached to it have been changed and the code has been revised to meet the 6:1 gear ratio. The new timing belt has a 1/8" width and 120 teeth, so a new pulley with maximum belt width of 1/8" has ordered for use along with the timing belt. Now, the pulley is small in size and the belt will not be parallel with the plate below it and that may affect the function therefore, a potentiometer plug extension, which extends the potentiometer to connect and to rotate along with the pulley, has been machined. The issue being faced now is that the belt ordered is a bit larger than needed, and a solution to fix that without having to return the belt and wait for a new one to arrive is to machine a belt tensioning pulley to be plugged into the plate at a distance that will put the belt in tension and rotates along with the belt.

Emergency Stop
A plate and a rectangular bar has been welded together to make the stand for the emergency stop and a box to hold the emergency stop has been 3D printed.

T-Plate Harness Attachment
A T-plate that attaches to the harnesses is missing therefore, a new one has been machined at the machine shop, shown below.

Wiring/Connections
To understand the circuitry of the apparatus and to know where and how to connect additional devices, all of the circuits has been disconnected then, rewired from scratch and that helped the team modify the wiring as needed for making the Twister more user-friendly specially, for the ease of movement of the whole apparatus from one location to another. Moreover, a different set of electrical connectors and plugs have been used to be able to disconnect/unplug most of the wires easily, when the Twisters needs to be disassembled for moving it,and to prevent reconnecting/replugging the wires the wrong way by using male and female plugs. The blue connectors plug into each other, the plugs with white heads plug into each other, while the emergency stop plug is of a bigger size than the previous ones therefore, they cannot be mistaken.

String Potentiometers
Instead of the 3D printed string potentiometers, two 10k Ohms string potentiometers have been ordered and installed on the lateral bar attachments for the sake of durability and the accuracy of measurements being acquired. The string potentiometers are connected to the 5V power supply as well as a BNC cable that goes to the computer. The potentiometers have been tested using an Oscilloscope and it displayed accurate measurements of displacement.

Next Semester's Goals
The team's goals for next semester are going to be to:
 * Work on making a flexible neck bar by installing U-joints.
 * Improve flat spring design attached to harnesses.
 * Install second lateral bar attachment.
 * Build V-track roller system for lateral bar attachments.
 * Build a mechanical safety stop for maximum rotation.
 * Move the Twister to Dr.Cohen's lab for testing and acquiring data.
 * Test torque sensors with Dr.Cohen's computer using the interface installed there.
 * Any other demands by client.