Crumbler Head Temperature Sensor
The purpose of this project is to develop a method of monitoring the surface temperature of a biomass grinder drum to notify the operator when product could potentially become damaged.
|Team Name||Team CrumbleTech|
|Duration||Fall 2017 - Spring 2018|
|Faculty Advisors||Dev Shrestha|
- 1 Problem Definition
- 2 Project Learning
- 3 Design Concept
- 4 Team Information
- 5 Document Archive
Forest Concepts LLC. is a company that manufactures and maintains Biomass grinders for use around the country. Biomass is produced by taking chips of a variety of materials that can range from wood chips to corn stalks and running them through a grinder to reduce the chips into a size that can be rapidly burned for a heat source. Forest Concepts came to the University of Idaho seeking a design that could measure the surface temperature of the two rotating drums that grind up the material to monitor for overheating which could potentially damage their product.
Forest Concepts has asked for the following deliverables:
- A reliable method of measuring temperature to ensure the product is not damaged
- A cheap sensor array for a permanent fixture that can be implemented on all crumblers or
- A removable sensor array that can be easily swapped between grinders if need be
- A sensor output that is readable by industry standard PLC's
- A method of verifying the temperature reading is accurate
After communicating further with Forest Concepts, the following specifications were determined:
- Maximum allowable temperature of 200 degrees Celsius
- Common running temperature of 100 – 150 Degrees Celsius
- Accuracy of +/- 10 -15 Degrees Celsius
- 4-20mA industry standard for Sensors
- 0-5 DC volts works for our operating system with their PLC
- The main products are Wood Chips, and Corn Stock
- The ash from these products is usually less than 3%
- Crumblerhead is made of A2 Tool Steel
- The teeth of the Crumblerhead are a Carbide Tip
- Must have a method for the verification of effectiveness of our Temperature Sensor System
- Prototype Sensor system will be tested on Forest Concepts Research Head
Emissivity is the ratio of energy radiated from a materials surface to that radiated from a black body. So, a black body would have an emissivity of 1.
Using the Stefan-Boltzmann law, we can determine the energy emitted by our drums as:
Where M is the flux per unit area of the source, 𝜀 is the emissivity of the object, 𝜎 is the Stefan-Boltzmann constant (5.67∗10−8𝑊𝑚−2𝐾−4) and T is the surface temperature of the object.
From this equation, we can derive the temperature for our infrared sensor which is called a Lambertian Source.
This sensor treats the entire surface area of the drum as a plane. Our sensor calculates the radius of the surface area projection with a constant Spot Ratio of 1:1 (1cm away = 1cm diameter of circle).
For our IR sensor type we can actually adjust the emissivity to match that of the material that the IR waves are being pointed at. The only major factor that could hinder our project would be the possibility of dust in between the IR and the crumblerhead.
Common values of emissivity based on other materials include:
- Oxidized steel - .79
- Silicon Carbide - .83-.96
- Gravel - .28
Thermocouples are composed of two dissimilar metals the generate an electric potential as one wire heats up faster than the other one based off of the material properties. For our experiment, a K-type thermocouple would most likely be the choice for measuring the temperature of the outgoing product if the customer so decides to go forward with that plan.
Degredation of Product
The entire reason of monitoring the crumbler head temperature is to ensure that ForestConcepts product is not being damaged by overheating. The three main components of wood that can actually melt if raised to roughly 450 Fahrenheit are Lignin, Cellulose, and Cellulose-lignin. If the temperature of the lignin is raised to this threshold, the normally-rigid protein will actually behave as a more fluid medium, causing permanent structural damage to the product as well as a loss of energy content in the burning during biomass fueling.
#1: IR Sensors
Design Idea: With each crumblerhead there are two cylindrical teethed rotating drums. So, for each of these drums we would have 5 IR sensors of Forest Concepts approval which would be evenly spaced out lengthwise along the drum. Each of these sensors will pick up a temperature for each section of the drum and whichever sensor is reading the highest temperature of the 5 will be displayed ion the PLC of the crumblerhead. The 5 sensors will be housed on the outside of the metal plates surrounding the crumbler head on either side of the drums for a total of 10 sensors per crumbler.
- Is a Permanent fixture to the system
- Cheap/easy maintenance
- Accurate to Forest Concepts wanted Specs
- Simple and effective
- Possible errors when high ash in-between the sensors and the drums
- Could have coding errors when trying to create functions to show max temperature with the sensor array
#2 Thermocouple on end product
Design Idea: At the end of the conveyor belt where the product is eventually drop into a container or bailer, we would have a thermocouple rod that as the product falls by it or passes by it on the conveyor it makes contact and the thermocouple would readout that temperature to the PLC.
- Directly measuring the product rather than the crumblerhead drum
- Simple to setup and install
- Is a permanent fixture on the system
- Hard to make sure your temperature readout function is accurate
- The time between the product leaving the crumblerhead and where the thermocouple is at may give the product time to cool down, providing inaccuracy.
- Possibly not very durable
For our design, we have chosen to use the infrared sensor arrays on both drums of the crumbler. These arrays will consist of 6 sensors per side for a total of 12 sensors per unit.
Part of the dilemma with the environment in which our sensor will be working is the amount of moving parts and particulate debris that will be circulating around our field of view. With that being said, we needed a sensor that was:
- rugged enough for use without maintenance
- fixed to the existing system
- easily inserted with minor modifications to existing side plate
The initial housing design consisted of a rapid-prototype housing with flanges to be inserted into a keyway in the side plate.
However, after further discussion with our clients, keyways are much harder to machine and much more expensive so we proceeded with the route of making the sensor cylindrical with a fixture on the outside of the plate. We have decided to fix the sensor housing to the plate wall via magnets due to the fact that they will not come loose with time as well as getting an IR-visible Lexan cover to keep dust out of our sensor housing.
Preliminary Data Acquisition
In order to being determining the emissivity of our sensor, ForestConcepts has asked us to create a Labview Program for use with our sensors. This can easily be done using an Arduino to control and gather information and use a serial communication with Labview in order to decipher the data. In order to do this, the file:Arduino sensor code.pdf must first be loaded into the arduino. This code cycles between one sensor every 1.5 seconds and is configurable for any 3 pins for Clock, Data, and Acquire pins.
Next, the Labview VI must be constructed to communicate with the Arduino via the VISA add-in. The back window looks as so:
Which yields the following front panel:
|Mechanical Engineering, Senior||I am from Boise, Idaho and I enjoy outdoor activities such as fly fishing, hunting, and hiking. In the future, I would like to work on R&D and product development.|
|Electrical Engineering, Senior||I am an electrical engineering student from china. I become a senior student at university of idaho recently. This is my first year in idaho. People in there are very friendly. I feel fresh and curious about the surrounding. My career goal is to create something which is useful and interesting.|
|Mechanical Engineering, Senior||My name is Abdulaziz Alazemi and I am from Kuwait. I am studying Mechanical Engineering at University of Idaho. About my achievements, I had learned English only in one year and started studying my Bachelor's degree in 2014. In addition, I had successful classes that made me a good and creative and capable to handle complex engineering problems. My hobbies are fishing, painting, cooking and swimming. When I graduate I will go back to Kuwait and hopefully be a member of Kuwait Water and Electricity Company.|
|Biological Engineering, Senior||My name is Anthony Ponzini and I am a Senior Biological Engineering student. Some of my hobbies include soccer, Guitar, video games, physical fitness, and eating different types of food. My last summer I interned for a company called Glanbia in southern Idaho and it was my first on the job engineering experience. I will be graduating this May and hope to have a bright future ahead of me.|